Siemens Symphony Magnetom 1.5T MRI 2004
Medrad Spectris Solaris Injector 3004213 11384
Series 23449 Software NUMARIS/4 Version syngo MR A40 DHHS Software COEM VD31H Syngo Vx91C SL02P08 N4_VA40A_LATEST_20130302 N4 SECURITY_UPDATE_0024_20140505 N4_VA40A_FEATURE_20130302_SP01_F10 N4_VA40A_FEATURE_201130302_OU2_F2 MR_MR018-17S_N4SeU0028 MR_MR023-17-R_N4SecU0029 Cryo 44% Coldhead
Siemens Primus LINAC 2002 3413
Low Photon Energy MV 6 Dose Rate MU/min 50&300 High Photon Energy MV 10,15,23 Dose Rate MU/min 10,15,23 Electron Energy MeV 6 to 21 Electron Dose Rate MU/min 300 or 900 Available with MLC Optional Minimum Room Dimensions: Depth 19' Width 18' Height 9'6" Power Requirements KVA 9'6" Water Requirements Gal/min 9'6"
The MRI is fully operational and with all supporting equipment. The LINAC is not operational and would either be for parts or for full refu rbishment.
The MRI Machine is on the third floor of a medical office building in Cente r City Philadelphia, Pennsylvania, USA and would have to be craned out thro ugh an outside wall. All costs for same would be by the taker, who (which) would have to be fully insured, licensed in Philadelphia, and bear all cost s of disassembly, removal, and restoration of building integrity. There wou ld be no direct costs for the machine or supporting equipment.
The LINAC machine is on the first floor of the same building, and subject t o the same requirements for removal - whatever means and methods and restor ation on the remover.
Please e-mail me directly. Either we get this done before 1/31/2020, or the machines both go out as scrap. Not kidding. Even a little bit.
Peter Wieck Melrose Park, PA