Do you have a question? Post it now! No Registration Necessary
May 4, 2005, 11:01 pm
rate this thread
We have a 8 year old 4 channel 2 Gsamples/sec 500MHz HP Infinium that has
suffered overvoltage in 2 of the 4 channels. Wondering if anyone has
attempted repair at component level themselves for this type of occurence?
Agilent will charge at least $3000 per channel for repair. Since 2 channels
are buggered and 1 is showing poor cal, it is time to decide to turf it and
get a Tektronix or attempt repair. Would like to here if anyone has done
- » Jak bezprzewodowo przekazać sygnał z sensora be zpieczeństwa?
- — The site's Last Updated Thread. Posted in » Electronics (Polish)