Do you have a question? Post it now! No Registration Necessary
- Posted on
- Attn DaveC re ESR meter
- Bob Parker
May 16, 2007, 3:30 pm
rate this thread
which have been trying to destroy this newsgroup, so my filters have
been deleting your posting(s). I only saw what you wrote by chance.
It would be preferable to discuss your ESR meter problems by direct
e-mail. You can contact me through this temporary e-mail address:
Re: Attn DaveC re ESR meter
Just a quick follow-up... there's probably nothing wrong with your
I hope you haven't wasted too much of your time chasing this
'fault'. When the 78L05 regulator is replaced with a LP2950CZ-5.0
and R26 is changed, the self-test function falsely shows an "F2" fault code.
Unfortunately this wasn't realized until after a lot of Mk2 kits
went out, because very few people make that modification. I've had a
warning about this on my website's ESR meter hints page for quite a long
If your meter's reading the 5.6 ohm and 68 ohm test resistors
reasonably accurately and doing all the things it's supposed to do,
there's nothing wrong with it. If it still has a problem, please e-mail
me at firstname.lastname@example.org
Sorry for the inconvenience this has caused. :(
BTW, C10 is 0.47uF not 47uF.
- » LOL OTT The 15 Most Hilarious, Absurd And Brutal Rate My Teacher Comments Ever!
- — Newest thread in » Electronics Down Under